Effects of design parameters on sensitivity of microcantilever biosensors
Lim, Yang-Choon, Kouzani, Abbas Z., Duan, Wei and Kaynak, Akif 2010, Effects of design parameters on sensitivity of microcantilever biosensors, in IEEE/ICME 2010 : Proceedings of the 2010 International Conference on Complex Medical Engineering, IEEE, [Gold Coast, Queensland], pp. 177-181.
Microcantilever biosensors produce cantilever bending due to differential surface stress between upper and lower surfaces of the cantilever. The bending is associated with concentration of ligands and adsorbed ligand-receptor intermolecular forces. Sample volume sizes in clinical diagnostic applications are usually very minute requiring a highly sensitive microcantilever for disease detection. This paper investigates a number of parameters that influence the sensitivity of microcantilever biosensors. The parameters include length, thickness, shape, and material of the cantilever beam. Biosensors of varying parameters are modeled and simulated. The results show that increasing the length of the cantilever beam enhances its sensitivity. However, increasing the thickness of the cantilever beam reduces its sensitivity. In static analysis, the shape of the cantilever beam does not notably impact upon its sensitivity. Also, using materials with lower Young’s modulus improves the sensitivity.
This material is presented to ensure timely dissemination of scholarly and technical work. Copyright and all rights therein are retained by authors or by other copyright holders. All persons copying this information are expected to adhere to the terms and constraints invoked by each author's copyright. In most cases, these works may not be reposted without the explicit permission of the copyright holder.
Field of Research
091306 Microelectromechanical Systems (MEMS) 090304 Medical Devices 090601 Circuits and Systems
Unless expressly stated otherwise, the copyright for items in DRO is owned by the author, with all rights reserved.
Every reasonable effort has been made to ensure that permission has been obtained for items included in DRO.
If you believe that your rights have been infringed by this repository, please contact email@example.com.
Every reasonable effort has been made to ensure that permission has been obtained for items included in DRO. If you believe that your rights have been infringed by this repository, please contact firstname.lastname@example.org.